VIDEOS 1 TO 50

Fabry Perot Interferometer

Published: 2016/10/23

Channel: JFC UCL

VTT Fabry-Perot interferometer technologies

Published: 2012/09/26

Channel: VTTFinland

Aligning a Scanning Fabry-Perot Interferometer

Published: 2016/01/26

Channel: Manoj Peiris

Fabry Perot

Published: 2014/10/21

Channel: Pat Kohl

Recording #5

Published: 2016/07/21

Channel: Sourabh Roy

Building of a Fabry-Perot Interferometer

Published: 2013/01/04

Channel: toc1955

Interferometer Fabry-Perot Sederhana

Published: 2015/04/27

Channel: Fisikaku Kakiku

(10) fabry perot interferometer شرح تاني

Published: 2016/04/24

Channel: Ahmed mohamed Srhan

Experimental Realization of a Fabry-Perot-Type Interferometer

Published: 2011/07/10

Channel: TAUVOD

What is Fabry-Perot FP Laser

Published: 2011/10/13

Channel: Colin Yao

Fabry-Perot Etalon

Published: 2013/11/12

Channel: BYU Physics & Astronomy

Rana Adhikari - Optics of Fabry-Perot cavities (Matrix approach for interferometer fields)

Published: 2014/02/21

Channel: International Centre for Theoretical Sciences

Homemade Fabry Perot interferometer

Published: 2012/10/23

Channel: toc1955

Helium Spectral Lines with a Fabry-Perot Interferometer

Published: 2012/11/14

Channel: toc1955

Microwave Fabry-Perot interferometer

Published: 2009/12/18

Channel: KoonPhysics

EM to Optics 31 : Resolving Power Fabry Perot Interferometer

Published: 2011/09/22

Channel: Adam Beatty

Rana Adhikari - Optics of Fabry-Perot cavities

Published: 2014/02/21

Channel: International Centre for Theoretical Sciences

Fabry Perot Experiment

Published: 2012/10/29

Channel: theSKYBLUETIES

252 - Fiber optic interferometer Fabry-Perot.

Published: 2013/05/27

Channel: Alexander C

Våglära och Optik, Lösningstips, Fabry-Perot Interferometer

Published: 2015/08/10

Channel: Waves and Optics

The FABRY-PEROT interferometer (Ανάλυση φωτος laser HeNe με φασματογράφο FABRY-PEROT)

Published: 2014/03/12

Channel: FunScience

Fabry--Pérot Interferometer The Movie CSUF

Published: 2011/03/15

Channel: Carrot Flowers

Fabry Perot Interferometer FP-28 sales@dmphotonics.com

Published: 2014/11/14

Channel: delmarphotonics

C7:Fabry-perot interferometer

Published: 2015/05/03

Channel: Ahmed Abd EL-Raheam

Interféromètre de Fabry Perot

Published: 2009/03/28

Channel: Gerald VINCENT

Fabry Perot Submission 1

Published: 2014/04/11

Channel: Jburrell01videos

Fabry-Perot Interferometer and Rings

Published: 2017/01/13

Channel: wolframmathematica

Etalon Meaning

Published: 2015/04/29

Channel: ADictionary

"New Advances in Fabry-perot Cavity for Sensing Applications " Dr. Orlando Frazão (PHOTOPTICS 2015)

Published: 2015/05/26

Channel: INSTICC

إيجاد الطول المولى لموجات الميكرويف (شرح احمد عبد الرحيم) (10) fabry perot interferometer

Published: 2016/04/24

Channel: Ahmed mohamed Srhan

Optical beat note between diode lasers locked to ultra stable Fabry Perot resonators

Published: 2011/01/21

Channel: minhenes

Individual or Small Team- Runner-Up: A potable Fabry-Perot Interferometer in your cellular phone.

Published: 2012/12/10

Channel: OpticalSociety

Introduction to VTT MEMS hyperspectral camera demo

Published: 2014/09/30

Channel: VTTFinland

Simple Fabry-Perot Etalon

Published: 2012/11/10

Channel: toc1955

Peter R Saulson - Optics of Fabry-Perot cavities (Plane wave picture, static...)

Published: 2014/02/21

Channel: International Centre for Theoretical Sciences

Lecture 33 Double Slit Interference; Interferometers

Published: 2014/06/04

Channel: tawkaw OpenCourseWare

205 - Surface analysis by fiber optic interferometer Fabry-Perot.

Published: 2013/05/26

Channel: Alexander C

Zeeman-effect in Fabry-Perot

Published: 2015/11/10

Channel: Pieter Kuiper

Fabry Perot tunable filter (FPTF) tuning

Published: 2014/04/11

Channel: Adrian Chance

beautiful fabry-perot fringes (by saeed abdee)

Published: 2014/10/12

Channel: Saeed Abdee

Fabry Perot pneumatic test hd

Published: 2011/06/07

Channel: Sergio Baron

Fabry Perot Cavity SAMPLE

Published: 2014/06/21

Channel: Peter Beyersdorf

Fabry Perot - Helium

Published: 2010/07/13

Channel: eladekel

Fabry Perot tunable filter (FPTF) tuning

Published: 2014/04/11

Channel: G&H Instruments

Fabry Perot pneumatic

Published: 2011/06/07

Channel: Sergio Baron

Fabry Perot - Mercury (Hg) Zeeman effect with filter

Published: 2010/07/13

Channel: eladekel

Fabry-Perot images

Published: 2012/10/24

Channel: toc1955

Fabry–Pérot etalons for single-mode lasers sales@dmphotonics.com

Published: 2014/12/27

Channel: delmarphotonics

FABRY-PEROT QUANTUM DOT LASERS - EXPERIMENTAL OBSERVATION OF EMISSION SPECTRUM

Published: 2012/05/30

Channel: Optcomm

PHYWE: Zeeman effect

Published: 2015/09/30

Channel: PHYWE Systeme GmbH & Co. KG

From Wikipedia, the free encyclopedia

"Étalon" redirects here. For the French commune, see Étalon, Somme. For the reference standard, see Standard (metrology).

This article needs additional citations for verification. (May 2016) (Learn how and when to remove this template message) |

In optics, a **Fabry–Pérot interferometer** or **etalon** is typically made of a transparent plate with two reflecting surfaces, or two parallel highly reflecting mirrors. (Precisely, the former is an etalon and the latter is an interferometer, but the terminology is often used inconsistently.) Its transmission spectrum as a function of wavelength exhibits peaks of large transmission corresponding to resonances of the etalon. It is named after Charles Fabry and Alfred Perot, who developed the instrument in 1899.^{[1]}^{[2]}^{[3]} *Etalon* is from the French *étalon*, meaning "measuring gauge" or "standard".^{[4]}

Etalons are widely used in telecommunications, lasers and spectroscopy to control and measure the wavelengths of light. Recent advances in fabrication technique allow the creation of very precise tunable Fabry–Pérot interferometers.

- 1 Basic description
- 2 Applications
- 3 Theory
- 3.1 1. Resonator losses, outcoupled light, resonance frequencies, and spectral line shapes
- 3.2 2. Generic Airy distribution: The internal resonance enhancement factor
- 3.3 3. Other Airy distributions
- 3.4 4. Airy distribution as a sum of mode profiles
- 3.5 5. Characterizing the Fabry-Pérot resonator: Lorentzian linewidth and finesse
- 3.6 6. Scanning the Fabry-Pérot resonator: Airy linewidth and finesse
- 3.7 7. Frequency-dependent mirror reflectivities
- 3.8 8. Description of the Fabry-Perot resonator in wavelength space
- 3.9 Detailed analysis
- 3.10 Another expression for the transmission function

- 4 See also
- 5 Notes
- 6 References
- 7 External links

The heart of the Fabry–Pérot interferometer is a pair of partially reflective glass optical flats spaced micrometers to centimeters apart, with the reflective surfaces facing each other. (Alternatively, a Fabry–Pérot *etalon* uses a single plate with two parallel reflecting surfaces.) The flats in an interferometer are often made in a wedge shape to prevent the rear surfaces from producing interference fringes; the rear surfaces often also have an anti-reflective coating.

In a typical system, illumination is provided by a diffuse source set at the focal plane of a collimating lens. A focusing lens after the pair of flats would produce an inverted image of the source if the flats were not present; all light emitted from a point on the source is focused to a single point in the system's image plane. In the accompanying illustration, only one ray emitted from point A on the source is traced. As the ray passes through the paired flats, it is multiply reflected to produce multiple transmitted rays which are collected by the focusing lens and brought to point A' on the screen. The complete interference pattern takes the appearance of a set of concentric rings. The sharpness of the rings depends on the reflectivity of the flats. If the reflectivity is high, resulting in a high Q factor, monochromatic light produces a set of narrow bright rings against a dark background. A Fabry–Pérot interferometer with high Q is said to have high *finesse*.

- Telecommunications networks employing wavelength division multiplexing have add-drop multiplexers with banks of miniature tuned fused silica or diamond etalons. These are small iridescent cubes about 2 mm on a side, mounted in small high-precision racks. The materials are chosen to maintain stable mirror-to-mirror distances, and to keep stable frequencies even when the temperature varies. Diamond is preferred because it has greater heat conduction and still has a low coefficient of expansion. In 2005, some telecommunications equipment companies began using solid etalons that are themselves optical fibers. This eliminates most mounting, alignment and cooling difficulties.
- Dichroic filters are made by depositing a series of etalonic layers on an optical surface by vapor deposition. These optical filters usually have more exact reflective and pass bands than absorptive filters. When properly designed, they run cooler than absorptive filters because they can reflect unwanted wavelengths. Dichroic filters are widely used in optical equipment such as light sources, cameras, astronomical equipment, and laser systems.
- Optical wavemeters and some optical spectrum analyzers use Fabry–Pérot interferometers with different free spectral ranges to determine the wavelength of light with great precision.
- Laser resonators are often described as Fabry–Pérot resonators, although for many types of laser the reflectivity of one mirror is close to 100%, making it more similar to a Gires–Tournois interferometer. Semiconductor diode lasers sometimes use a true Fabry–Pérot geometry, due to the difficulty of coating the end facets of the chip. Quantum cascade lasers often employ Fabry-Pérot cavities to sustain lasing without the need for any facet coatings, due to the high gain of the active region.
^{[5]} - Etalons are often placed inside the laser resonator when constructing single-mode lasers. Without an etalon, a laser will generally produce light over a wavelength range corresponding to a number of cavity modes, which are similar to Fabry–Pérot modes. Inserting an etalon into the laser cavity, with well-chosen finesse and free-spectral range, can suppress all cavity modes except for one, thus changing the operation of the laser from multi-mode to single-mode.
- Fabry–Pérot etalons can be used to prolong the interaction length in laser absorption spectrometry, particularly cavity ring-down, techniques.
- A Fabry–Pérot etalon can be used to make a spectrometer capable of observing the Zeeman effect, where the spectral lines are far too close together to distinguish with a normal spectrometer.
- In astronomy an etalon is used to select a single atomic transition for imaging. The most common is the H-alpha line of the sun. The Ca-K line from the sun is also commonly imaged using etalons.
- In gravitational wave detection, a Fabry–Pérot cavity is used to
*store*photons for almost a millisecond while they bounce up and down between the mirrors. This increases the time a gravitational wave can interact with the light, which results in a better sensitivity at low frequencies. This principle is used by detectors such as LIGO and Virgo, which consist of a Michelson interferometer with a Fabry–Pérot cavity with a length of several kilometers in both arms. Smaller cavities, usually called*mode cleaners*, are used for spatial filtering and frequency stabilization of the main laser.

The spectral response of a Fabry-Pérot resonator is based on interference between the light launched into it and the light circulating in the resonator. Constructive interference occurs if the two beams are in phase, leading to resonant enhancement of light inside the resonator. If the two beams are out of phase, only a small portion of the launched light is stored inside the resonator. The transmitted and reflected light is spectrally modified compared to the incident light.

Assume a two-mirror Fabry-Pérot resonator of geometrical length , homogeneously filled with a medium of refractive index . Light is launched into the resonator under normal incidence. The round-trip time of light travelling in the resonator with speed , where is the speed of light in vacuum, and the free spectral range are given by

The electric-field and intensity reflectivities and , respectively, at mirror are

If there are no other resonator losses, the photon-decay time of the resonator is then given by^{[6]}

With quantifying the single-pass phase shift that light exhibits when propagating from one mirror to the other, the round-trip phase shift at frequency accumulates to^{[6]}

Resonances occur at frequencies at which light exhibits constructive interference after one round trip. Each resonator mode with its mode index , where is an integer number in the interval [, …, −1, 0, 1, …, ], is associated with a resonance frequency and wavenumber ,

Two modes with opposite values and of modal index and wavenumber, respectively, physically representing opposite propagation directions, occur at the same absolute value of frequency.

The decaying electric field at frequency is represented by a damped harmonic oscillation with an initial amplitude of and a decay-time constant of . In phasor notation, it can be expressed as^{[6]}

Fourier transformation of the electric field in time provides the electric field per unit frequency interval,

Each mode has a normalized spectral line shape per unit frequency interval given by

whose frequency integral is unity. Introducing the full-width-at-half-maximum (FWHM) linewidth of the Lorentzian spectral line shape, we obtain

Calibrated to a peak height of unity, we obtain the Lorentzian lines:

When repeating the above Fourier transformation for all the modes with mode index in the resonator, one obtains the full mode spectrum of the resonator.

Since the linewidth and the free spectral range are independent of frequency, whereas in wavelength space the linewidth cannot be properly defined and the free spectral range depends on wavelength, and since the resonance frequencies scale proportional to frequency, the spectral response of a Fabry-Pérot resonator is naturally analyzed and displayed in frequency space.

The response of the Fabry-Pérot resonator is most easily derived by use of the circulating-field approach.^{[7]} This approach assumes a steady state and relates the various electric fields to each other (see figure to the right).

The field can be related to the field that is launched into the resonator by

The generic Airy distribution, which considers solely the physical processes exhibited by light inside the resonator, then derives as the intensity circulating in the resonator relative to the intensity launched,^{[6]}

represents the spectrally dependent internal resonance enhancement which the resonator provides to the light launched into it (see figure to the right). At the resonance frequencies , where equals zero, the internal resonance enhancement factor is

Once the internal resonance enhancement, the generic Airy distribution, is established, all other Airy distributions can be deduced by simple scaling factors.^{[6]} Since the intensity launched into the resonator equals the transmitted fraction of the intensity incident upon mirror 1,

and the intensities transmitted through mirror 2, reflected at mirror 2, and transmitted through mirror 1 are the transmitted and reflected/transmitted fractions of the intensity circulating inside the resonator,

respectively, the other Airy distributions with respect to launched intensity and with respect to incident intensity are^{[6]}

The index “emit” denotes Airy distributions that consider the sum of intensities emitted on both sides of the resonator.

The back-transmitted intensity cannot be measured, because also the initially back-reflected light adds to the backward-propagating signal. The measurable case of the intensity resulting from the interference of both backward-propagating electric fields results in the Airy distribution^{[6]}

It can be easily shown that in a Fabry-Pérot resonator, despite the occurrence of constructive and destructive interference, energy is conserved at all frequencies:

The external resonance enhancement factor (see figure to the right) is^{[6]}

At the resonance frequencies , where equals zero, the external resonance enhancement factor is

Usually light is transmitted through a Fabry-Pérot resonator. Therefore, an often applied Airy distribution is^{[6]}

It describes the fraction of the intensity of a light source incident upon mirror 1 that is transmitted through mirror 2 (see figure to the right). Its peak value at the resonance frequencies is

For the peak value equals unity, i.e., all light incident upon the resonator is transmitted; consequently, no light is reflected, , as a result of destructive interference between the fields and .

has been derived in the circulating-field approach^{[7]} by considering an additional phase shift of during each transmission through a mirror,

resulting in

Alternatively, can be obtained via the round-trip-decay approach^{[8]} by tracing the infinite number of round trips that the incident electric field exhibits after entering the resonator and accumulating the electric field transmitted in all round trips. The field transmitted after the first propagation and the smaller and smaller fields transmitted after each consecutive propagation through the resonator are

respectively. Exploiting

results in the same as above, therefore the same Airy distribution derives.

Physically, the Airy distribution is the sum of mode profiles of the longitudinal resonator modes.^{[6]} Starting from the electric field circulating inside the resonator, one considers the exponential decay in time of this field through both mirrors of the resonator, Fourier transforms it to frequency space to obtain the normalized spectral line shapes , divides it by the round-trip time to account for how the total circulating electric-field intensity is longitudinally distributed in the resonator and coupled out per unit time, resulting in the emitted mode profiles,

and then sums over the emitted mode profiles of all longitudinal modes^{[6]}

thus equaling the Airy distribution .

The same simple scaling factors that provide the relations between the individual Airy distributions also provide the relations among and the other mode profiles:^{[6]}

The Taylor criterion of spectral resolution proposes that two spectral lines can be resolved if the individual lines cross at half intensity. When launching light into the Fabry-Pérot resonator, by measuring the Airy distribution, one can derive the total loss of the Fabry-Pérot resonator via recalculating the Lorentzian linewidth , displayed (blue line) relative to the free spectral range in the figure below.

The underlying Lorentzian lines can be resolved as long as the Taylor criterion is obeyed (see figure to the right on the physical meaning of the Lorentzian finesse). Consequently, one can define the Lorentzian finesse of a Fabry-Pérot resonator:^{[6]}

It is displayed as the blue line in the figure to the right. The Lorentzian finesse has a fundamental physical meaning: it describes how well the Lorentzian lines underlying the Airy distribution can be resolved when measuring the Airy distribution. At the point where

equivalent to , the Taylor criterion for the spectral resolution of a single Airy distribution is reached. For equal mirror reflectivities, this point occurs when . Therefore, the linewidth of the Lorentzian lines underlying the Airy distribution of a Fabry-Pérot resonator can be resolved by measuring the Airy distribution, hence its resonator losses can be spectroscopically determined, until this point.

When the Fabry-Pérot resonator is used as a scanning interferometer, i.e., at varying resonator length (or angle of incidence), one can spectroscopically distinguish spectral lines at different frequencies within one free spectral range. Several Airy distributions , each one created by an individual spectral line, must be resolved. Therefore, the Airy distribution becomes the underlying fundamental function and the measurement delivers a sum of Airy distributions. The parameters that properly quantify this situation are the Airy linewidth and the Airy finesse . The FWHM linewidth of the Airy distribution is^{[6]}

The Airy linewidth is displayed as the green curve in the figure on linewidths and finesses above.

The concept of defining the linewidth of the Airy peaks as FWHM breaks down at (solid red line in the figure on ), because at this point the Airy linewidth instantaneously jumps to an infinite value. For lower reflectivity values , the FWHM linewidth of the Airy peaks is undefined. The limiting case occurs at

For equal mirror reflectivities, this point is reached when (solid red line in the figure on ).

The finesse of the Airy distribution of a Fabry-Pérot resonator, which is displayed as the green curve in the figure above in direct comparison with the Lorentzian finesse , is defined as^{[6]}

When scanning the length of the Fabry-Pérot resonator (or the angle of incident light), the Airy finesse quantifies the maximum number of Airy distributions created by light at individual frequencies within the free spectral range of the Fabry-Pérot resonator, whose adjacent peaks can be unambiguously distinguished spectroscopically, i.e., they do not overlap at their FWHM (see figure to the right on the physical meaning of the Airy finesse). This definition of the Airy finesse is consistent with the Taylor criterion of the resolution of a spectrometer. Since the concept of the FWHM linewidth breaks down at , consequently the Airy finesse is defined only until , see the figure above.

Often the unnecessary approximation is made when deriving from the Airy linewidth . In contrast to the exact solution above, it leads to

This approximation of the Airy linewidth, displayed as the red curve in the figure above, deviates from the correct curve at low reflectivities and incorrectly does not break down when . This approximation is then typically also used to calculate the Airy finesse.

The more general case of a Fabry-Pérot resonator with frequency-dependent mirror reflectivities can be treated with the same equations as above, except that the photon decay time and linewidth now become local functions of frequency. Also in this case each Airy distribution is the sum of all underlying mode profiles which can be strongly distorted.^{[6]} An example of the Airy distribution and a few of the underlying mode profiles is given in the figure to the right.

The varying transmission function of an etalon is caused by interference between the multiple reflections of light between the two reflecting surfaces. Constructive interference occurs if the transmitted beams are in phase, and this corresponds to a high-transmission peak of the etalon. If the transmitted beams are out-of-phase, destructive interference occurs and this corresponds to a transmission minimum. Whether the multiply reflected beams are in phase or not depends on the wavelength (λ) of the light (in vacuum), the angle the light travels through the etalon (θ), the thickness of the etalon (*ℓ*) and the refractive index of the material between the reflecting surfaces (*n*).

The phase difference between each successive transmitted pair (i.e. T2 - T1 in the diagram) is given by δ:^{[9]}

If both surfaces have a reflectance *R*, the transmittance function of the etalon is given by

where

is the *coefficient of finesse*.

Maximum transmission () occurs when the optical path length difference () between each transmitted beam is an integer multiple of the wavelength. In the absence of absorption, the reflectance of the etalon *R*_{e} is the complement of the transmittance, such that . The maximum reflectivity is given by:

and this occurs when the path-length difference is equal to half an odd multiple of the wavelength.

The wavelength separation between adjacent transmission peaks is called the free spectral range (FSR) of the etalon, Δλ, and is given by:

where λ_{0} is the central wavelength of the nearest transmission peak and is the group refractive index.^{[citation needed]} The FSR is related to the full-width half-maximum, δλ, of any one transmission band by a quantity known as the *finesse*:

- .

This is commonly approximated (for *R* > 0.5) by

- .

If the two mirrors are not equal, the finesse becomes

- .

Etalons with high finesse show sharper transmission peaks with lower minimum transmission coefficients. In the oblique incidence case, the finesse will depend on the polarization state of the beam, since the value of "R", given by the Fresnel equations, is generally different for p and s polarizations.

A Fabry–Pérot interferometer differs from a Fabry–Pérot etalon in the fact that the distance *ℓ* between the plates can be tuned in order to change the wavelengths at which transmission peaks occur in the interferometer. Due to the angle dependence of the transmission, the peaks can also be shifted by rotating the etalon with respect to the beam.

Fabry–Pérot interferometers or etalons are used in optical modems, spectroscopy, lasers, and astronomy.

A related device is the Gires–Tournois etalon.

This section does not cite any sources. (October 2011) (Learn how and when to remove this template message) |

Two beams are shown in the diagram at the right, one of which (T_{0}) is transmitted through the etalon, and the other of which (T_{1}) is reflected twice before being transmitted. At each reflection, the amplitude is reduced by , while at each transmission through an interface the amplitude is reduced by . Assuming no absorption, conservation of energy requires *T* + *R* = 1. In the derivation below, *n* is the index of refraction inside the etalon, and *n*_{0} is that outside the etalon. It is presumed that *n* > *n*_{0}. The incident amplitude at point a is taken to be one, and phasors are used to represent the amplitude of the radiation. The transmitted amplitude at point b will then be

- ,

where is the wavenumber inside the etalon and λ is the vacuum wavelength. At point c the transmitted amplitude will be

- .

The total amplitude of both beams will be the sum of the amplitudes of the two beams measured along a line perpendicular to the direction of the beam. The amplitude *t*_{0} at point b can therefore be added to *t'*_{1} retarded in phase by an amount where is the wavenumber outside of the etalon. Thus

- ,

where ℓ_{0} is

- .

The phase difference between the two beams is

- .

The relationship between *θ* and *θ*_{0} is given by Snell's law:

- ,

so that the phase difference may be written:

- .

To within a constant multiplicative phase factor, the amplitude of the *m*th transmitted beam can be written as:

- .

The total transmitted amplitude is the sum of all individual beams' amplitudes:

The series is a geometric series whose sum can be expressed analytically. The amplitude can be rewritten as

- .

The intensity of the beam will be just *t* times its complex conjugate. Since the incident beam was assumed to have an intensity of one, this will also give the transmission function:

For an asymmetrical cavity, that is, one with two different mirrors, the general form of the transmission function is

Defining the above expression may be written as:

The second term is proportional to a wrapped Lorentzian distribution so that the transmission function may be written as a series of Lorentzian functions:

where

- Lummer–Gehrcke interferometer
- Atomic line filter
- ARROW waveguide
- Distributed Bragg reflector
- Fiber Bragg grating
- Optical microcavity
- Thin-film interference

**^**Perot frequently spelled his name with an accent—Pérot—in scientific publications, and so the name of the interferometer is commonly written with the accent. Métivier, Françoise (September–October 2006). "Jean-Baptiste Alfred Perot" (pdf).*Photoniques*(in French) (25). Retrieved 2007-10-02. Page 2: "Pérot ou Perot?"**^**Fabry, C; Perot, A (1899). "Theorie et applications d'une nouvelle methode de spectroscopie interferentielle".*Ann. Chim. Phys*.**16**(7).**^**Perot, A; Fabry, C (1899). "On the Application of Interference Phenomena to the Solution of Various Problems of Spectroscopy and Metrology".*Astrophysical Journal*.**9**.**^***Oxford English Dictionary***^**Williams, Benjamin S. (2007). "Terahertz quantum-cascade lasers".*Nature Photonics*.**1**(9): 517–525. doi:10.1038/nphoton.2007.166. ISSN 1749-4885.- ^
^{a}^{b}^{c}^{d}^{e}^{f}^{g}^{h}^{i}^{j}^{k}^{l}^{m}^{n}^{o}^{p}^{q}^{r}^{s}^{t}^{u}^{v}^{w}N. Ismail, C. C. Kores, D. Geskus, and M. Pollnau, "Fabry-Pérot resonator: spectral line shapes, generic and related Airy distributions, linewidths, finesses, and performance at low or frequency-dependent reflectivity", Optics Express 24 (15), 16366-16389 (2016). doi:10.1364/OE.24.016366. - ^
^{a}^{b}A. E. Siegman, "Lasers", University Science Books, Mill Valley, California, 1986, ch. 11.3, pp. 413-428. **^**O. Svelto, "Principles of Lasers", 5th ed., Springer, New York, 2010, ch. 4.5.1, pp. 142-146.**^**Lipson, S.G.; Lipson, H.; Tannhauser, D.S. (1995).*Optical Physics*(3rd ed.). London: Cambridge U.P. p. 248. ISBN 0-521-06926-2.

- Hernandez, G. (1986).
*Fabry–Pérot Interferometers*. Cambridge: Cambridge University Press. ISBN 0-521-32238-3.

- Compact FP interferometer for gas analysis
- Advanced Design of Etalons- by Precision Photonics Corporation

Wikipedia content is licensed under the GFDL and (CC) license