Polarimetry is the measurement and interpretation of the polarization of transverse waves, most notably electromagnetic waves, such as radio or light waves. Typically polarimetry is done on electromagnetic waves that have traveled through or have been reflected, refracted or diffracted by some material in order to characterize that object.
Polarimetry of thin films and surfaces is commonly known as ellipsometry.
Polarimetry can also be included in computational analysis of waves. For example, radars often consider wave polarization in post-processing to improve the characterization of the targets. In this case, polarimetry can be used to estimate the fine texture of a material, help resolve the orientation of small structures in the target, and, when circularly-polarized antennas are used, resolve the number of bounces of the received signal (the chirality of circularly polarized waves alternates with each reflection).
A polarimeter is the basic scientific instrument used to make these measurements, although this term is rarely used to describe a polarimetry process performed by a computer, such as is done in polarimetric synthetic aperture radar.
Polarimetry can be used to measure various optical properties of a material, including linear birefringence, circular birefringence (also known as optical rotation or optical rotary dispersion), linear dichroism, circular dichroism and scattering. To measure these various properties, there have been many designs of polarimeters, some archaic and some in current use. The most sensitive are based on interferometers, while more conventional polarimeters are based on arrangements of polarising filters, wave plates or other devices.
Polarimetry is used in many areas of astronomy to study physical characteristics of sources including active galactic nuclei and blazars, exoplanets, gas and dust in the interstellar medium, supernovae, gamma-ray bursts, and the cosmic microwave background radiation. Astronomical polarimetry observations are carried out either as imaging polarimetry, where polarization is measured as a function of position in imaging data, or spectropolarimetry, where polarization is measured as a function of wavelength of light.
Optically active samples, such as solutions of chiral molecules, often exhibit circular birefringence. Circular birefringence causes rotation of the polarization of plane polarized light as it passes through the sample.
In ordinary light, the vibrations occur in all planes perpendicular to the direction of propagation. When light passes through a Nicol prism its vibrations in all directions except the direction of axis of the prism are cut off. The light emerging from the prism is said to be plane polarised because its vibration is in one direction. If two Nicol prisms are placed with their polarization planes parallel to each other, then the light rays emerging out of the first prism will enter the second prism. As a result, no loss of light is observed. However, if the second prism is rotated by an angle of 90°, the light emerging from the first prism is stopped by the second prism and no light emerges. The first prism is usually called the polarizer and the second prism is called the analyser.
A simple polarimeter to measure this rotation consists of a long tube with flat glass ends, into which the sample is placed. At each end of the tube is a Nicol prism or other polarizer. Light is shone through the tube, and the prism at the other end, attached to an eye-piece, is rotated to arrive at the region of complete brightness or that of half-dark, half-bright or that of complete darkness. The angle of rotation is then read from a scale. The same phenomenon is observed after an angle of 180°. The specific rotation of the sample may then be calculated. Temperature can affect the rotation of light, which should be accounted for in the calculations.