A spectrum analyzer measures the magnitude of an input signal versus frequency within the full frequency range of the instrument. The primary use is to measure the power of the spectrum of known and unknown signals. The input signal a spectrum analyzer measures is electrical, however, spectral compositions of other signals, such as acoustic pressure waves and optical light waves, can be considered through the use of an appropriate transducer.
By analyzing the spectra of electrical signals, dominant frequency, power, distortion, harmonics, bandwidth, and other spectral components of a signal can be observed that are not easily detectable in time domain waveforms. These parameters are useful in the characterization of electronic devices, such as wireless transmitters.
The display of a spectrum analyzer has frequency on the horizontal axis and the amplitude displayed on the vertical axis. To the casual observer, a spectrum analyzer looks like an oscilloscope and, in fact, some lab instruments can function either as an oscilloscope or a spectrum analyzer.
|This section requires expansion. (December 2012)|
The first spectrum analyzers, in the 1960s, were swept-tuned instruments.
Today, there are three basic types of analyzer: the swept-tuned spectrum analyzer, the vector signal analyzer, and the real-time spectrum analyzer.
Spectrum analyzer types are dictated by the methods used to obtain the spectrum of a signal. There are swept-tuned and FFT based spectrum analyzers:
Some spectrum analyzers, such as real-time spectrum analyzers, use a hybrid technique where the incoming signal is first down-converted to a lower frequency using superheterodyne techniques and then analyzed using fast fourier transformation (FFT) techniques.
Spectrum analyzers tend to fall into three form factors: benchtop, portable and handheld.
This form factor is useful for applications where the spectrum analyzer can be plugged into AC power, which generally means in a lab environment or production/manufacturing area. Bench top spectrum analyzers have historically offered better performance and specifications than the portable or handheld form factor. Bench top spectrum analyzers normally have multiple fans (with associated vents) to dissipate heat produced by the processor. Due to their architecture, bench top spectrum analyzers typically weigh more than 30 pounds (14 kg). Some bench top spectrum analyzers offer optional battery packs, allowing them to be used away from AC power. This type of analyzer is often referred to as a "portable" spectrum analyzer.
This form factor is useful for any applications where the spectrum analyzer needs to be taken outside to make measurements or simply carried while in use. Attributes that contribute to a useful portable spectrum analyzer include:
This form factor is useful for any application where the spectrum analyzer needs to be very light and small. Handheld analyzers offer a limited capability relative to larger systems. Attributes that contribute to a useful handheld spectrum analyzer include:
As discussed above in types, a swept-tuned spectrum analyzer down-converts a portion of the input signal spectrum to the center frequency of a band-pass filter by sweeping the voltage-controlled oscillator through a range of frequencies, enabling the consideration of the full frequency range of the instrument.
The bandwidth of the band-pass filter dictates the resolution bandwidth, which is related to the minimum bandwidth detectable by the instrument. As demonstrated by the animation to the right, the smaller the bandwidth, the more spectral resolution. However, there is a trade-off between how quickly the display can update the full frequency span under consideration and the frequency resolution, which is relevant for distinguishing frequency components that are close together. For a swept-tuned architecture, this relation for sweep time is useful:
Where ST is sweep time in seconds, k is proportionality constant, Span is the frequency range under consideration in Hertz, and RBW is the resolution bandwidth in Hertz. Sweeping too fast, however, causes a drop in displayed amplitude and a shift in the displayed frequency.
Also, the animation contains both up- and down-converted spectra, which is due to a frequency mixer producing both sum and difference frequencies. The local oscillator feedthrough is due to the imperfect isolation from the IF signal path in the mixer.
With a FFT based spectrum analyzer, the frequency resolution is , the inverse of the time T over which the waveform is measured and Fourier transformed.
With Fourier transform analysis in a digital spectrum analyzer, it is necessary to sample the input signal with a sampling frequency that is at least twice the bandwidth of the signal, due to the Nyquist limit. A Fourier transform will then produce a spectrum containing all frequencies from zero to . This can place considerable demands on the required analog-to-digital converter and processing power for the Fourier transform, making FFT based spectrum analyzers limited in frequency range.
Since FFT based analyzers are only capable of considering narrow bands, one technique is to combine swept and FFT analysis for consideration of wide and narrow spans. This technique allows for faster sweep time.
This method is made possible by first down converting the signal, then digitizing the intermediate frequency and using superheterdyne or FFT techniques to acquire the spectrum.
One benefit of digitizing the intermediate frequency is the ability to use digital filters, which have a range of advantages over analog filters such as near perfect shape factors and improved filter settling time. Also, for consideration of narrow spans, the FFT can be used to increase sweep time without distorting the displayed spectrum.
Most modern spectrum analyzers are now almost exclusively Hybrid Superheterodyne-FFT based giving a significant improvement in sweep time. However, even in such cases there is still processing time required to sample the spectrum and calculate the FFT. For this reason, both swept-tuned and FFT based analyser produce "blind time" meaning that while calculation of the spectrum is being performed, the instrument has gaps and misses information of the RF spectrum being applied to the RF front end.
A realtime spectrum analyser does not have any such blind time--up to some maximum span, often called the "realtime bandwidth". The analyser is able to sample the incoming RF spectrum in the time domain and convert the information to the frequency domain using the FFT process. FFT's are processed in parallel, gapless and overlapped so there are no gaps in the calculated RF spectrum and no information is missed.
In a sense, any spectrum analyzer that has vector signal analyzer capability is a realtime analyzer. It samples data fast enough to satisfy Nyquist Sampling theorem and stores the data in memory for later processing. This kind of analyser is only realtime for the amount of data / capture time it can store in memory and still produces gaps in the spectrum and results during processing time.
In all spectrum analyzers we want to have information with the least distortion possible. The FFT process applies windowing techniques to improve the output spectrum due to producing less side lobes. The effect of windowing may also reduce the level of a signal where it is captured on the boundary between one FFT and the next. For this reason FFT's in a Realtime spectrum analyzer are overlapped. Overlapping rate is approximately 80%. An analyzer that utilises a 1024 point FFT process will re-use approximately 819 samples from the previous FFT process.
This is related to the sampling rate of the analyser and the FFT rate. It is also important for the realtime spectrum analyzer to give good level accuracy.
Example: for an analyser with 40 MHz of realtime bandwidth (the maximum RF span that can be processed in realtime) approximately 50 Msample/second (complex) are needed. If the spectrum analyzer produces 250 000 FFT/s an FFT calculation is produced every 4 µs. For a 1024 point FFT a full spectrum is produced 1024 x (1/50 x 106), approximately every 20 µs. This also gives us our overlap rate of 80% (20 µs − 4 µs) / 20 µs = 80%.
Realtime spectrum analyzers are able to produce much more information for users to examine the frequency spectrum in more detail. A normal swept spectrum analyzer would produce max peak, min peak displays for example but a realtime spectrum analyzer is able to plot all calculated FFT's over a given period of time with the added colour-coding which represents how often a signal appears. For example, this image shows the difference between how a spectrum is displayed in a normal swept spectrum view and using a "Persistence" view on a realtime spectrum analyzer.
Realtime spectrum analyzers are able to see signals hidden behind other signals. This is possible because no information is missed and the display to the user is the output of FFT calculations. An example of this can be seen on the right.
In a typical spectrum analyzer there are options to set the start, stop, and center frequency. The frequency halfway between the stop and start frequencies on a spectrum analyzer display is known as the center frequency. This is the frequency that is in the middle of the display’s frequency axis. Span specifies the range between the start and stop frequencies. These two parameters allow for adjustment of the display within the frequency range of the instrument to enhance visibility of the spectrum measured.
As discussed in the operation section, the resolution bandwidth filter or RBW filter is the bandpass filter in the IF path. It's the bandwidth of the RF chain before the detector (power measurement device). It determines the RF noise floor and how close two signals can be and still be resolved by the analyzer into two separate peaks. Adjusting the bandwidth of this filter allows for the discrimination of signals with closely spaced frequency components, while also changing the measured noise floor. Decreasing the bandwidth of an RBW filter decreases the measured noise floor and vice versa. This is due to higher RBW filters passing more frequency components through to the envelope detector than lower bandwidth RBW filters, therefore a higher RBW causes a higher measured noise floor.
The video bandwidth filter or VBW filter is the low-pass filter directly after the envelope detector. It's the bandwidth of the signal chain after the detector. Averaging or peak detection then refers to how the digital storage portion of the device records samples—it takes several samples per time step and stores only one sample, either the average of the samples or the highest one. The video bandwidth determines the capability to discriminate between two different power levels. This is because a narrower VBW will remove noise in the detector output. This filter is used to “smooth” the display by removing noise from the envelope. Similar to the RBW, the VBW affects the sweep time of the display if the VBW is less than the RBW. If VBW is less than RBW, this relation for sweep time is useful:
Where ST is sweep time in seconds, k is proportionality constant, Span is the frequency range under consideration in Hertz, RBW is the resolution bandwidth in Hertz, and VBW is the video bandwidth in Hertz.
With the advent of digitally based displays, some modern spectrum analyzers use analog-to-digital converters to sample spectrum amplitude after the VBW filter. Since displays have a discrete number of points, the frequency span measured is also digitised. Detectors are used in an attempt to adequately map the correct signal power to the appropriate frequency point on the display. There are in general three types of detectors: sample, peak, and average
The Displayed Average Noise Level (DANL) is just what it says it is--the average noise level displayed on the analyzer. This can either be with a specific resolution bandwidth (usually in dBm), or normalized to 1 Hz (usually in dBm/Hz)
In telecommunications, spectrum analyzers are used to determine occupied bandwidth and track interference sources. For example, cell planners use this equipment to determine interference sources in the GSM frequency bands and UMTS frequency bands.
In EMC testing, a spectrum analyzer is used for basic precompliance testing; however, it can not be used for full testing and certification. Instead, an EMI receiver like the Rohde & Schwarz ESU EMI Receiver, Agilent Technologies N9038A MXE EMI, or Gauss Instruments TDEMI is used.
A spectrum analyzer is used to determine whether a wireless transmitter is working according to federally defined standards for purity of emissions. Output signals at frequencies other than the intended communications frequency appear as vertical lines (pips) on the display. A spectrum analyzer is also used to determine, by direct observation, the bandwidth of a digital or analog signal.
A spectrum analyzer interface is a device that connects to a wireless receiver or a personal computer to allow visual detection and analysis of electromagnetic signals over a defined band of frequencies. This is called panoramic reception and it is used to determine the frequencies of sources of interference to wireless networking equipment, such as Wi-Fi and wireless routers.
Spectrum analyzers can also be used to assess RF shielding. RF shielding is of particular importance for the siting of a magnetic resonance imaging machine since stray RF fields would result in artifacts in an MR image.
Spectrum analysis can be used at audio frequencies to analyse the harmonics of an audio signal. A typical application is to measure the distortion of a nominally sinewave signal; a very-low-distortion sinewave is used as the input to equipment under test, and a spectrum analyser can examine the output, which will have added distortion products, and determine the percentage distortion at each harmonic of the fundamental. Such analysers were at one time described as "wave analysers". Analysis can be carried out by a general-purpose digital computer with a sound card selected for suitable performance and appropriate software. Instead of using a low-distortion sinewave, the input can be subtracted from the output, attenuated and phase-corrected, to give only the added distortion and noise, which can be analysed.
An alternative technique, total harmonic distortion measurement, cancels out the fundamental with a notch filter and measures the total remaining signal, which is total harmonic distortion plus noise; it does not give the harmonic-by-harmonic detail of an analyser.
|Wikimedia Commons has media related to: Spectrum analyzers|
Here you can share your comments or contribute with more information, content, resources or links about this topic.